Indexing Mount for Ø1" Optics
Through Hole for
These indexing mounts allow a shared laser beam to be redirected within multiple setups or for an optical signal to be interchanged between multiple test instruments. The NX1N(/M) indexing optic mount is designed to hold Ø1" optics no more than 7 mm (0.27") thick at one of 16 discrete positions, while the NX1F(/M) indexing mounting base features 15 discrete positions as well as a top-located 8-32 (M4) tapped hole for general-purpose mounting. Both indexing mounts have three 8-32 (M4) tapped mounting holes on the bottom plate, enabling multiple mounting options. In addition, a through hole in the side of the base accepts a 5/64" (2.00 mm) balldriver to attach the mount to a standard Ø1/2" post.
Please Note: The top plate of the NX1N(/M) indexing mount for Ø1" optics is not compatible with the NX1F(/M) indexing mounting base.
A selection of Thorlabs Kinematic Bases were tested to determine the deviation of a reflected beam attributed to removing and replacing the top plate of the base. The test setup (see photo below and on the left) consisted of an S1FC635 fiber-coupled laser source, terminating in a fiber collimator that was held in a POLARIS-K1 Mirror Mount. The collimated beam was directed towards a BB1-E02 mirror housed in an LMR1 fixed mount that was post mounted on the kinematic base being tested (see photo below and center). After reflecting off the first mirror, the beam traveled towards and reflected off of a second mirror housed in a POLARIS-K1 mount. After reflection, the light was directed onto a PDP90A Lateral Effect Position Sensor that had three SM05L10 SM05-Threaded Lens Tubes attached to the front to block stray light (see photo below and to the right). The total beam path is shown in the first photo and is 1 m long.
The test was conducted by first centering the beam on the sensor through adjustment of the Polaris mount holding the second mirror. After centering the beam, thirty seconds were allowed to pass for stabilization purposes, and then a measurement of the beam position was made and recorded as the inital position. Next, the top plate of the kinematic base was lifted from the bottom and then replaced in the same orientation. After a thirty second stabilization period, a second measurement of the beam position was taken. Finally, the Polaris mirror mount was readjusted to center the beam on the sensor, and the process was repeated. Sixty data points were collected, amounting to thirty measurements of the beam displacement.
The results of the measurements are presented in the table and graphs below as the absolute difference in µrad between the original position of the beam on the sensor and the position after removing and replacing the top plate of the kinematic base. Each value has been divided by two to account for the reflection off the second mirror. The combined value takes into account both X- and Y-position measurements and represents the net distance of the deviation from the origin.
Click to Enlarge
The NX1N(/M) Indexing Mount can hold an optic at any one of 16 positions, including 45° and 90°.
The NX1N(/M) indexing mount for Ø1" optics features 16 indexed positions every 22.5°, allowing the optic to be held at both 45° and 90°. The angular positions are engraved on the base plate for fast alignment. The top plate returns to any given position, indicated with the engraved witness line, with a repeatability of better than 10 µrad. This mount is ideal for redirecting a laser to multiple targets on an optical table.
The included SM1 (1.035"-40) retaining ring (SM1RR) secures the optic against the front lip of the mount. This indexing mount can hold optics that are no greater than 7 mm (0.27") thick. Neodymium iron magnets couple the top and bottom plates, while stainless steel ball bearings and V-grooves provide the 16 indexed positions. Three 8-32 (M4) tapped mounting holes in the bottom plate allow the mount to be secured to a Ø1/2" post. A small through hole, located directly under the 90° engraving on the bottom plate, accepts a 5/64" (2.00 mm) balldriver to help provide enough torque when attaching the mount to the post.
Please Note: The top plate of this indexing mount is not compatible with the NX1F(/M) 15-position, indexing mounting base sold below.
The NX1F(/M) mounting base is an indexed platform that accepts a wide variety of optomechanical accessories such as kinematic mirror mounts. The top platform provides a center-located 8-32 (M4) tapped hole for general-purpose mounting. The kinematic ball-seat design provides 15 discrete positions and repeatability within 10 µrad. Each position is laser engraved with a capital letter, from A through O, for fast placement.
Please Note: The NX1F(/M) indexing mounting base is not compatible with the NX1N(/M) indexing Ø1" optic mount (sold above).